"Now, without taking anybody out of service, we can fix a magnetized CT in less than 1 minute!"
CT Demagnetization Circuit--600-DMG1
Before Demag--Ratio 212.4
After Demag--Ratio 200.2
of you have seen a measured ratio test reading of a CT that was not what it should
example, a 200:5 CT that measured out to be 212:5 but passed the burden test.
This is a magnetized CT.
And what can you do about it?
Either nothing or change out the CT.
Now there is a 3rd option:
The Spinlab Demag Circuit used in tandem with the Bird Dog Plus.
Of the 32 failure modes of a meter circuit, magnetized CTs are by far the
most common ones after installation errors.
A magnetized CT can be caused by a lightning strike, ground fault, or
open circuiting a CT.
With this shoebox size device, you simply attach the duckbill connector
to the phase of the test switch you want to demagnetize, turn the unit on, press
the Demag button, and in less than 1 minute your CT will be demagnetized.
It is that easy, and you don’t have to take anything out of service!
Never has increasing the accuracy of a CT-metered site been easier.
Here’s more about the Demag circuit:
Works on any CT-rated site—primary or secondary
on secondary current from 0.2 to 10 amps
lockouts built in for excessive current, voltage, and temperature readings
directly to the Bird Dog Plus for easy before and after monitoring to prove
size and weight allows you to carry it easily from site to site
predetermines the extent of the test needed to demagnetize a CT based on
load and degree of magnetization
a CT normally in less than 1 minute
batteries to maintain—works from site power
a Printable Demagnetization Circuit Flyer in .pdf Format Click Here)
To place an Order or to request a Price Quotation
contact our Sales Force at 865 212 9881
Home | About Spinlab | Why Test? | 5000 Bird Dog Plus | 6000 Bird Dog Plus | Load Box | Accessories | Literature | Customer Service
Spinlab Utility Instrumentation, Inc. / 3110 Henson Road, Suite A1 / Knoxville, TN 37921
© 2008, Spinlab Utility Instrumentation, Inc.